Position your bracelet and adjust the tightness
Place the bracelet on the top, flat side of your wrist and on the elbow side of your wrist bone.
The sensor is in contact with your skin. The bracelet fits snugly but it is not uncomfortable.
If you are having troubles with the initialization process and have seen the error message signaling “poor signal quality”, please try the following:
1. Take off the bracelet
2. Rub your wrist’s skin for around 10 seconds to improve skin perfusion
3. Tighten your bracelet again so that it is well in contact with your skin
Place your cuff and adjust the tightness
Remove the garments from your upper arm. Do not roll up your sleeve as this may cause blood flow constriction and affect the quality of the measurement. Tighten your Aktiia cuff on your left arm, the blue LED and Aktiia logo should be close to your elbow joint.
The cuff measurement body is aligned with your brachial artery and sits 2-3 cm above your elbow joint.
The cuff should fit snugly around your arm but you should be able to slide two fingers between your arm and the cuff.
Adjust your posture for the measurement
During the measurement, your entire forearm, from hand to elbow, is resting on a flat surface.
Make sure your forearm is NOT leaning against the table edge to avoid constriction on the brachial artery.
Seated, lean your back against the chair in a relaxed but straight position. Both your feet should be on the ground, your knees forming a 90 degree angle.
Both forearms are resting on a flat surface.
The bracelet and the cuff are on opposite arms.
Before starting the initialization process please relax in a seated position for 5 minutes.
To ensure a successful initialization please:
• Do not perform physical activities during the 30 minutes preceding the measurement;
• Do not drink caffeinated beverages during the 30 minutes preceding the measurement;
• Do not smoke during the 30 minutes preceding the measurement;
• Do not speak or move during the measurement.
Please note that sometimes, despite having followed all the steps carefully, a successful measurement cannot be performed.
This may be due to low skin perfusion at your wrist or to your blood pressure varying too much within the measurements. In these cases, Aktiia app will ask you to quit the measurement and come back in few hours, to ensure the accuracy of the measurement.
It is important to follow the instruction and wait at least 2 hours before trying again.
In case you are still having issues with the process after 3 attempts performed at different times of the day, please reach out to our customer support at firstname.lastname@example.org – we’ll gladly provide more guidance.